Last update: February 1998
Anorad system should now be locked up and ready to operate.
Turn on Air to table (toggle valve on green tank, below the delay line
tank, next to the Fluor table).
Check air flow rate to values specified on card - about 5 liters/minute
with air filled tank, 4 liters/minute with vacuum. Higher flow rates will
induce oscillations of the carriage location and
orientation. Valves need to be adjusted when changing between vacuum and
air filled tanks.
Have Rate cards powered up and a delay line control program operating.
Command the delay lines to be idle (stopped).
Turn on Anorad Servo (upper blue box on the big rack).
Clear laser faults by pushing reset (front black button on the big rack)
on HP box.
Back to Instrument Subsystems Page